|Date Added:||18 November 2005|
|File Size:||17.87 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Message 8 of 9. Message 2 of 9. Helee 11 1 2.
Sign up using Email and Password. Message 5 of 9. Message 6 of 9. I am interested in doing something very similar.
There should be an example with the driver to do what you want. In the end, I used the KI driver supplied by Keithley. Most Active Software Boards: Message 9 of 9.
Email Required, but never shown. I am using GPIB controller to connect Keithley to my PC, however I don’t have any experience using this labview software, can anyone suggest how to create this kind of program?
Keithley 2401 Low Voltage SMU SourceMeter, 20V, 1A, 20W
I am currently studying how to control the voltage and the current of a source meter which is Keithley using Labview software. How to use labview to control the voltage and current of Keithley Ask Question.
I’m just navigating a path based on that email I received from NI driver support group that I pasted in my last comment, the forums I’ve read etc. DakotaD 1 Sign up or log in Sign up using Google.
Keithley Low Voltage SMU SourceMeter, 20V, 1A, 20W
Hey, if downloading, installing, rebooting, debugging, uninstalling, repairing etc etc to get the software to work properly didn’t take me have a week, I’d probably just try it out.
Of course its always possible to simply open them up and do just that. Anyway the image linked here shows how you could source and measure keitbley using KE and how to read out and parse the data returned.
Sign up using Facebook. Joe Friedrichsen 1, 12 Message 7 of 9.